BTEX and 1,3 Butadiene analyzer: airTOXICBTXPID
A GC/PID for automatic monitoring of BTEX in air, water and soils
This instrument uses a port valve with a single absorbent trap, and a 30 meter metallic 0.28 mm ID column.
Minimum detection level as low as 10 ppt for application Benzene or 1,3-Butadiene in ambient air.
Miniaturization, sensitivity, mobility and flexibility are its main features.
Everything from the sample port up to the data storage is integrated in a 19"- rack 5U.
MCERTS certified analyser for benzene measurement following EN 14662-3 norms
Programmed gradient temperature up to 202°c for the column oven and pressure/flow control of the carrier gas .
Linearity on area with R2 > 0.995
European range : 0.45 to 45 µg/m3 = 0.1 to 14.5 ppb.
Before delivery the analyser is tested for one week by our quality control department.
Photo ionisation detector (PID), 10,6 eV lamp.
Self-cleaning of the lamp ensures high stability.
Bi-directionnal RS-232 C to transfer data and results to the internal computer.
Vistachrom software enables the user to view and store data on a industrial computer. It provides comfortable user friendly utilities to recalculate, calibrate and export data and to configure the measurement.
The software allows the calculation of retention time, area, mass or concentration profiles, in any measuring unit.
Results output : different communication protocols available
airTOXIC BTX certificate
TSP airTOXIC BTX
Back to previous page
BTEX and 1,3 Butadiene analyzer: airTOXICBTXPID download
US EPA report (430.2 kb)
LCSQA report (1.5 Mb)
Related Applications :
Our airToxic analyzer has been certified in 2006 by a European notified body, CNR for BTEX detection with a PID detector.
This market is driven by law and legislations made in Europe and/or USA (US EPA). It deals with ambient air monitoring : analysis / calibration / automatic zero air in continuous 7 days a week. (...)
Chromatotec can propose for hazardous areas (for instance refineries) specific boxes for our analysers : category 1 IIc T4 for example.
In the United States and in Asia the classification is (...)
After the 2006 edition, this is the second time CHROMATOTEC participates to the ARAB LAB exhibition and this is with great pleasure we come back to Dubaï!
The Middle Eastern area is well known (...)
Some VOC’s like 1,3 butadiene or benzene are known to be carcinogenic.
The French ministerial recommendation (DGS / SD 7 B no 2005-273 of February 25th, 2005) relating to the evaluation of the (...)
US EPA has developed Methods TO-14 (39 compounds) and TO-15 (75 compounds) to measure Toxic Organics Air Pollutants. Sampling is done typically via canisters and then analyzed typically in the (...)
The US EPA has performed a 6 months comparative study of 1,3Butadiene and benzene in ambient air with our airToxic GC analyzer A76022 against the canister method TO-15 at Las Vegas, Nevada.
US EPA has performed a 6 months comparative study of 1,3Butadiene and benzene in ambient air with our airToxic GC analyzer A76022 against the canister method TO-15 at Las Vegas, Nevada.
Real time (...)
Chromatotec is pleased to announce that our airTOXIC PID analyzer has recently received another certification! On March 1st 2010 we received the Pattern Approval Certification according to the (...)
Chromatotec is certified with the European Standard EN-15267 for automated measuring systems manufacturing and EN 14662-3:2015 - standard method for the measurement of benzene (...)
Franck Amiet, Michel Robert, Dr Nicholas A Martin